- Call the patient's VAD coordinator ASAP
- These patients may not have a palpable pulse. Listen over the heart to hear if the motor is working. Then use mental status, skin color/temp, and the machine flashing Low Flow as indicators that perfusion badness is occurring. Do a bedside echo. The MAP should be ~65 on manual doppler BP, Automated BP devices may give you a MAP as well. A-line MAP is the most accurate.
- Try not to cut or yank out the drive-line, 'cause that is embarrassing.
- When in doubt, consider a fluid bolus. VADS love volume. If you need to improve hemodynamics with a working LVAD, consider preload augmentation and possibly afterload reduction (if MAP is high).
- Consider inotropes–if you think it is right heart failure, give dobutamine. If you think the patient is septic and has markedly reduced afterload, consider norepinephrine.
- On echo:
- Big RV, small LV=pulm hypertension or right heart strain/stemi. Correct hypoxemia and acidosis, consider volume, screen for RV STEMI, consider inotropes.
- Small RV-give volume
- Big RV & LV-pump failure or pump thrombosis.
Consider pump thrombosis–Signs of pump thrombosis are LVAD is hot, working hard, with high RPM, low flow, dilated RV/LV, and low MAP. Zack would give a bolus of 5000 U of Heparin in the decompensating LVAD that he thought was secondary to thrombosis (or if he just couldn't figure out what was wrong with a failing device). He would also consider tPA if he really thought it was pump thrombosis and the patient was decompensating and peri-code.
On ECHO, a dilated RV/LV could be from pump thrombosis or non-working pump (electrical issue for example).if you think that is the problem, heparinize.
Machine Not Running
Check batteries. Make sure all of the lines are connected.
These folks are prone to bleeding from the anticoag (and probably additional plt dysfunction from the device if I had to guess). So if they have altered mental status or neuro findings–consider hemorrhagic stroke.
Patient appears Infected
Drive-line infection-look at the site at entry to the skin. If the patient appears septic and you can't find a source, consider it a device infection until proven otherwise. Don't yank the device. Treat for health-care associated infection covering both hospital gram negatives and MRSA.
Patient is Coding
We need to AVOID CPR until the patient needs it and at point, what is the alternative? Can you rip out the device with CPR-yes! Many of the CT surgeons recommend not to do CPR, but you can't get deader than dead (I was not a philosophy major, so I could be wrong). Avoid CPR if at all possible, some of the 1st gen devices had hand-pumps you could use–the current generation don't. If you're the point where there is NOTHING else to do except CPR you need to use your clinical judgment.
Here is Zack's clinical judgment:
CPR is not recommended by the manufacturers secondary to potential cannula dislodgement. I would not do CPR unless the pump was NOT working and the patient had lost their BP (MAP of 0). This is the one scenario where you have to perfuse the brain no matter what the cost. All other scenarios I would focus on how to get that pump operating better (at all).
Joe Bellezo then adds:
Just agreeing with Zack's thoughts on this. My approach to this is ‘Look, Listen, and feel” – assuming a comatose LVAD patient.
Look: …at all the connections. Everything connected? Ok. Look at the controller. Green Light on? Yes..ok. No? Troubleshoot for a problem with the VAD and keep working until you get the green light on the controller.
Listen: to the hum. Assuming a green light on the controller…there should be an LVAD Hum. No hum? the pump isn't functional (duh). Find out why. Again, check all the connections and then touch the control box and check RPM, flow, etc). Pump thrombosis is your reversible problem here.
Feel: hot control box is not good and usually means thrombosis or dislodgement of the outflow cannula to the aorta…or distal obstruction like a dissection.
Compressions: here is my thought: if you've gone through all of the above and there is nothing to fix…then you have an LVAD patient who does not have a functional LVAD. I would treat them just as if they came to the ED the day before they got their LVAD: a patient with end stage heart failure and no blood pressure. I would begin chest compressions if their MAP was below 60 because they aren't perfusing their vital organs and will die. I know this goes against Zack's recommendations but that shows you that nobody really knows the best answer here. This patient will die. I say start the chest compressions and get inotropes going. Dobutamine or milrinone stat in addition to levophed. In other words…pretend they don't have a VAD and aggressively resuscitate them. Yes, dislodging the pump is possible…but these patients are going to die anyway.
Though if I had to guess Zack and Joe would only do CPR long enough to crash the patient on to ECMO.
Read this PDF Now
An insanely good field guide from mylvad.com.It has device-specific recommendations. Read it NOW!
The site also has some excellent additional resources.
Another great review on LVADs from Fire Engineering
the images from this post are from mylvad
Great Review (Emerg Med Austral 2014;26:104)
Now on to the Podcast…
Latest posts by Scott Weingart (see all)
- A Letter to CMS from Paul Marik - April 29, 2017
- Ketamine ……. then Rocuronium, DSI & The Timing Principle - April 25, 2017
- Podcast 197 – The Logistics of the Administration of Massive Transfusion - April 17, 2017